The Wrong Way To Program Hearing Aids | Functional Gain. Dr. Cliff Olson, Audiologist and founder of Applied Hearing Solutions in Anthem Arizona, discusses the wrong way to program hearing aids and what hearing care providers should do instead.
Some hearing care professionals still don’t verify if hearing aids are programmed correctly, and others are still using old fashioned methods of hearing aid verification that are outdated at best. This old fashioned method is called Functional Gain.Functional gain testing is when hearing aids are auto-programmed and adjusted for your hearing loss, and then you are taken into a sound booth to see if you can hear "beep" sounds at a softer level. This method of verifying hearing aid programming does not provide a full understanding of how a hearing aid is programmed for the following reasons:
1. Unless you want to hear Star Wars character R2D2 talk, then testing how well you hear "beeps" with your hearing aids is useless.
2. Functional gain doesn’t provide a good understanding of the level of amplification that you will receive in more realistic listening situations. Hearing aids are programmed for different input levels ranging from 55 dB, 65 dB, and 80 dB SPL. Testing beeps at levels lower than 55 dB will not indicate if hearing aids are programmed correctly for typical speech-level inputs.
3. Functional Gain doesn't help with precision programming of hearing aids. Functional gain is performed in 5 dB increments. Hearing aids can be adjusted in 1 dB increments. This means that the results of Functional gain testing won't even tell you how to adjust the hearing aids.
4. You may not get accurate readings using functional gain. Compression can impact the high-frequency readings while using functional gain. This means that it could show that the hearing aid isn't programmed correctly and cause an over-amplification of the high-frequencies.
All that being said, Functional Gain testing is probably better than not verifying hearing aid programming at all.
However, it would be even better to verify hearing aid programming using Real Ear Measurement. Real Ear Measurement is the Gold Standard of hearing aid verification. Without it, programming is complete guesswork. If you want to learn more about Real Ear Measurement, check out my video: https://youtu.be/cHR0Oa6I-wY
Hearing aid manufacturers have done an excellent job of making hearing aids smaller and smaller.Read More
The vast majority of hearing losses can be treated effectively with the use of hearing aids.Read More
Hearing aid users- this one is for you.Read More
Hearing aids come in various styles, each of which are appropriate for different types and severities of hearing loss.Read More
A quick look into over the counter hearing aids, what they are and what they do.Read More
Satisfaction with hearing instruments has steadily improved over the last 30 years.Read More